The composition of DR generally includes high-voltage generating port, X-ray tube and bracket, flat-panel detection port, system control port, etc. Compared with conventional X-ray signals, in addition to the advantages of CR, the DR system uses flat-panel detection. The X-ray receiving device replaces the traditional intensifying screen and film, realizes the digitization of X-ray signals, the dynamic range of the signal, the spatial resolution and density resolution are high, and the exposure agent is further reduced.
dr occasionally does not expose light: when the first gear of the DR exposure handbrake can be REAY pressed, it means that there is no problem with the high-voltage control part. After the second gear is pressed, the signal will send a signal to the CANON control part. If the signal cannot be transmitted to the CANON host or Can not be fed back from the CANON host, will cause the high-voltage part to be in a waiting state and cannot be exposed.
If you are interested in DR, you are welcome to contact us
Mobile DR X-ray Machine
Hot Line: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Add: E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China