DR prolonged downtime precautions.
Within 1 week of downtime
The probability that the tablet does not reach the normal working state increases, the probability that the generator does not occasionally reach the normal working state increases, the probability that the exposure is not synchronized increases, and the probability that the equipment accumulates dust increases.
Downtime for more than 7 days
The tablet needs to be powered on continuously for more than 4 hours to enter the working state.
The probability of tube training is increased, the flat panel has not reached the normal working condition, and the image quality has not reached the optimal level.
Downtime for more than 30 days
It is necessary to check the power supply environment of the whole machine before starting the machine, and to train the tube. The tablet needs to be continuously powered on for more than 4 hours, and calibration may be required. The generator needs to be calibrated according to the actual usage.
The use of environmental problems on the equipment, environmental temperature problems.
The operating temperature of the equipment room is lower than 10°, which causes the panel to be abnormal or damaged. The shutdown temperature of the equipment room is lower than 5° to cause irreversible damage to the panel. The room humidity or water ingress causes the panel’s life to drop sharply, and the dust in the room is too large. The life span of the whole machine is drastically reduced.
Electricity environmental issues
The input voltage is too high or too low to cause the whole machine to work abnormally, the input voltage of the grid is unstable and fluctuates to cause the whole machine board to be overloaded, the ground wire of the DR equipment fails, poor grounding causes equipment failure or personnel injury, and frequent power outages in hospitals cause equipment Unable to reach steady state of work.
Department environmental issues
The random entry and exit of non-departmental personnel caused the equipment to be artificially damaged,
The surrounding environment of the department causes small animals to enter and exit at will, causing equipment such as cables to be destroyed, not paying attention to the environmental hygiene of the department, failing to clean up food residues to lure mice and other small animals into, failing to turn on the air-conditioning equipment as required, resulting in the ambient temperature when restarting Unable to meet the requirements.
The impact of bad operating habits on equipment
Ignoring the ambient temperature and forcing the tablet to work below 10° caused the machine to frequently report the D23 error. If it is not turned on and warmed up for a period of time in a low temperature environment, the image quality will be poor, and the temperature will be below freezing for a long time to cause irreversible damage to the tablet.
Mobile DR X-ray Machine
Hot Line: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Add: No.1 Factory Building No.13426 Yuqing East Street, Weifang, Shandong, China