For the failure of the X-ray machine to not image, it can be checked from two aspects. First check whether the generator emits rays, if no rays are emitted, check the generator. For example, whether the handbrake switch is damaged, whether the generator is faulty (you can open the generator software separately to check), and whether there is a problem with the tube. The generator can emit rays normally, and the panel side should be checked. For example, whether the connection of the tablet is normal (whether the network cable or optical fiber is connected correctly), or whether the synchronization signal (exposure enable signal) of the tablet is normal. Looking for faults from these two aspects, you can find faults.
If you are interested in our X-ray machine, you are welcome to contact us. Phone +8617616362243!
Mobile DR X-ray Machine
Hot Line: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Add: No.1 Factory Building No.13426 Yuqing East Street, Weifang, Shandong, China