The process of using the DR device is as follows:
Boot sequence: power on the monitor, power on the imaging workstation, power on the X-ray machine, and then power on the post-processing workstation.
Shutdown sequence: Turn off the power of the X-ray machine, turn off the power of the imaging workstation, turn off the power of the post-processing workstation, and turn off the power of the monitor. When the workstation is turned off, click the lower right corner of the monitor to close the system and exit each application program until the display program is closed. At this time, the power of the host can be turned off. When closing the post-processing workstation, click the icon in the upper left corner of the display until the display program is closed, at which point the power can be turned off.
The precautions for DR devices are as follows:
When working, keep the indoor temperature at 18–25 degrees and the humidity at about 45%.
The DR equipment system should be used as slowly as possible to prevent damage to the mechanical parts.
There should be no heavy objects squeezed or impacted on the frame panel, so as not to damage the detector.
Shut down not too often to avoid damaging the machine.
Regularly calibrate the detector, it is recommended to calibrate the detector every four weeks.
Work strictly in accordance with the operating procedures, and notify our company in time when the machine encounters an obstacle.
If you want to know more about DR equipment, you are welcome to contact me at +8617616362243!
Mobile DR X-ray Machine
Hot Line: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Add: No.1 Factory Building No.13426 Yuqing East Street, Weifang, Shandong, China