X-ray machine does not produce images after exposure
This fault can be checked from two aspects. First check whether the generator emits rays, if no rays are emitted, then
Check the generator. For example, whether the handbrake switch is damaged, whether the generator is faulty (you can turn on the generator soft
Check the parts), whether there is a problem with the tube. If the generator can emit rays normally, the panel side should be checked. such as flat
Whether the connection of the board is normal (whether the network cable or optical fiber is connected correctly), or the synchronization signal of the board (exposure enable signal)
number) is normal. Looking for faults from these two aspects, you can find faults.
The image from the X-ray machine is not clear
Over time, the image becomes less clear than it used to be. The cause of this problem – generally due to the tablet
Caused by aging or insufficient dose from the bulb. Regular maintenance of the equipment can effectively alleviate the aging of the tablet,
Plates generally require calibration every 6 months. Causes of tube underdosing include:
(1) KV is not enough, check the part of the high voltage generator.
(2) Insufficient MA, whether there is a problem with the bulb, or whether there is a problem with the filament of the high-voltage generator.
Mobile DR X-ray Machine
Hot Line: +86 536-8882360
Company: Weifang Newheek Electronic Tech Co., Ltd.
Add: No.1 Factory Building No.13426 Yuqing East Street, Weifang, Shandong, China